Just a small concept I've built today and would like to share. I needed to measure a resonance of a LC tank inside a circuit.
It all took one RG-316 pigtail and 5 minutes of work. Results are satisfactory (apply you preferred disclaimers here).
Beware of cheap underperforming clones
As of 2022 there are many badly performing clones on the market. V2/3GHz NanoVNA uses parts like ADF4350 and AD8342 which are costly and clones have been cutting costs by using salvaged or reject parts.
See official store and look for V2 Plus4/V2 Plus4 Pro versions only to avoid getting a bad clone. We have stopped selling V2.2 versions since October 2020, so all V2 hardware that are not Plus or Plus4 are not made by us and we can not guarantee performance.
Click here to join and see most recent posts.
LC resonance "sonde"
I have one practical remark - as far as the NanoVNA has 50Ohm low impedance output, measurement of parallel LC circuit should be not done in series, but the best way is to couple to the parallel LC circuit inductively with a small loop or similar, such as when you measure with the GDO. By connecting to the parallel LC in this way the LC circuit is significantly loaded. The graph is unclear, what was measured- was it S11 or S21?
The measurement in this way is more suitable for series LC circuit. This is the theory.
Yes, I have used GDO-like loop (connected in series as S11) for other measurements, but inside a tight circuit it is physically impossible to get a reading this way. Thus I have crafted this tool. I calibrate these "tweezers" for trough and isolation, and in theory any pure-resistive loading (50R by nanoVNA) should not affect dip frequency.
Tweezers are just connected to both SMA ports, so I measure S21 loss - it is in the top left of the chart.
To reply to this topic, join https://groups.io/g/NanoVNAV2