Question: When calibrations are run they do for the full range 10k -3Ghz or just the selected current range ?
Beware of cheap underperforming clones
As of 2022 there are many badly performing clones on the market. V2/3GHz NanoVNA uses parts like ADF4350 and AD8342 which are costly and clones have been cutting costs by using salvaged or reject parts.
See official store and look for V2 Plus4/V2 Plus4 Pro versions only to avoid getting a bad clone. We have stopped selling V2.2 versions since October 2020, so all V2 hardware that are not Plus or Plus4 are not made by us and we can not guarantee performance.
Click here to join and see most recent posts.
Calibration is done for the selected frequency range, avg setting, number of
points, etc etc
That is the reason for having seven memory slots...
Am 14.12.2020 09:46 schrieb "Silviu Isari via groups.io"
> Question: When calibrations are run they do for the full range 10k -3Ghz or
just the selected current range ?
* * *
To reply to this topic, join https://groups.io/g/NanoVNAV2